Home Technical Documentation Piecewise Calibration  
See also: RecalcCalibPieces, RecalcCalibPieces


Piecewise Calibration 

The calibration model of ImageLab is quite flexible. For each axis ImageLab provides one or several calibration functions which map each index of the data array to the corresponding world coordinates. The calibration function can be any kind of polynomial function of orders zero to six, including piecewise calibration functions of different orders. Technical implementation: the metadata container maintains two individual polynomial functions for each index of the fourdimensional data cube, one function for the forward transform (converting indices to world coordinates) and one for the inverse transform (world coordinates to indices). This elaborate scaling scheme ensures that almost any kind of calibration can be achieved to a high level of precision. Please keep in mind, that the calibration functions may not necessarily differ from index to index, in fact, in many cases the calibration function is constant over all indices of a given axis (i.e. the lateral calibration is normally a linear one with constant parameters, since nonlinear calibration will distort the resulting image). Under practical circumstances the only calibration function which is more complicated than a simple linear relationship will occur along the spectral axis. For example, a data cube might contain three different kinds spectra (= spectral groups) which of course requires at least three different calibration functions. Another example for a more complex calibration function is the piecewise combination of various functions along the spectral axis (as it may occur with instruments containing several detectors). In order to achieve a fast response piecewise calibrations are recorded and maintained in an internal calibration table which contains the boundaries of the individual pieces of the calibration function. The figure above shows a calibration function which consists of five pieces, some of them being straight lines, others being polynomials. Please note that calibration functions  even piecewise ones  have to be monotonous with no gaps in between (i.e. there must be a valid forward and inverse function for each index of the axis and the indices of adjacent calibration function must be consecutive at the boundaries). The extended version of ImageLab provides several ILabPascal functions which support the calculation and management of calibration functions, i.e. GetCalibCoeffs, SetCalibCoeffs, RecalcCalibPieces and CalcInvCalibCoeffs.
